HS101-PRO is an upgrated version of the HS101 Oscilloscope that joins the portability of an Android oscilloscope with advanced hardware features. The build require some experience with SMT mounting so approachable just from part of the DIY community. Within its specs, all the standard oscilloscope applications are possible with this device.
±1.5V to ±15V in 4 steps
up to 12 Bits (effective without noise: 9 bit, 10bit @ 100KSa/s, 11bit @ 75KSa/s, 12bit @ 12KSa/s)
3KS/s to 1800KS/s in 13 steps
(sine wave -3dB)
Sine wave visible with sinc interpolation up to 400KHz @ 1800KS/s
Up to 100KSa/s
< 60mV (<= 15mV for Sampling Rate <= 100KSa/s)
<=20mV with the Black Pill (<=10mV for Sampling Rate <= 100KSa/s)
AC/DC (hardware implemented)
Modules supported in HScope
Automotive Module, Audio Module
1. How to build it
The hardware is based on STM32F103 microcontroller. It integrates 2 fast 12bit ADCs for the data acquisition. The Black Pill development board has been chosen for its low ADC noise among several development boards, so suiteable for this project.
Under this board we are going to install a little PCB module for input acquisition.
The input module includes the hardware AC/DC coupling and a variable signal amplifier with 1MΩ input. Here is the building scheme for this module.
List of the components:
|Microcontroller||STM32F103C8 Black Pill|
|Microcontroller Programming Tool||Raspberry Pi USB to TTL Cable (or another model specified here)|
|PCB Board||Custom PCB|
|SMT Components||All components in the schematic||USB Cable||USB OTG Cable - Micro USB to Micro USB or USB Type-C to Micro USB (according the device)|
First Release: May 2018
Last Update: Jun 2019